Objectives: The study compares the efficacy of two active and one passive warming interventions in healthy volunteers with induced mild hypothermia.
Methods: Eight volunteers were studied in a random order crossover design. Each volunteer was studied during re-warming from a core temperature of 35°C with each of: a radiant warmer (Fisher & Paykel); a forced air warmer (Augustine Medical), and a polyester filled blanket, to re-warm.
Results: No significant differences in re-warming rates were observed between the three warming devices. It was found that the subject’s endogenous heat production was the major contributor to the re-warming of these volunteers. Metabolic rates of over 350 W were seen during the study.
Conclusions: For patients with mild hypothermia and in whom shivering is not contraindicated our data would indicate that the rate of re-warming would be little different whether a blanket or one of the two active devices were used. In the field, this may provide the caregiver a useful choice.
- BMI, body mass index
- patient warming
- radiant heating
Statistics from Altmetric.com
If you wish to reuse any or all of this article please use the link below which will take you to the Copyright Clearance Center’s RightsLink service. You will be able to get a quick price and instant permission to reuse the content in many different ways.
Competing interests: three of the authors work for Fisher & Paykel Healthcare Corporation, manufacturer of one of the devices tested